Hot loop water for the bell jar is only at 27 C, setpoint is 35C. Need to order a new heater. Ran a pumping speed vs. time bell jar is open. When the bell jar has been open for 2 min., the pump down time is 2.9 min. When the bell jar open time = 10 min., the pump down time = 5.8 min.