[aw610_r] Need TC Mode for AW610R
rc3534 at stanford.edu
Tue Mar 12 21:29:01 PDT 2013
Is there any reason everything is still in pyro mode? My understanding is
that the tool is primarily in TC mode with day exceptions for the
pyrometer. If the TC setup and calibration are working, why don't we
install the TC back on after pyro testing is done for the day so that it's
not dead for everyone else?
-------------- next part --------------
An HTML attachment was scrubbed...
More information about the aw610_r