[cascade] Fwd: Re: Cascade lamp source repair update
shott at stanford.edu
Fri Jan 24 16:40:39 PST 2014
Jim McVittie has found a temporary replacement for the light source on
Cascade. To prevent damage from overheating, please note his comment
that you should leave the illuminator in the fan position for a couple
of minutes before turning it completely off.
Of course a work of thanks to Jim for a creative and effective solution
is always in order. Thanks, JIm.
-------- Original Message --------
Subject: Re: [cascade] Cascade lamp source repair update
Date: Fri, 24 Jan 2014 15:04:13 -0800 (PST)
From: Jim McVittie <mcvittie at stanford.edu>
To: Woo Shik Jung <alsune at stanford.edu>, cascade at snf.stanford.edu
CC: Jim McVittie <mcvittie at cis.stanford.edu>
I have borrowed a fiber illuminator from Tom Lee's lab and made an adapter for our fiber. So we now have a microscope light on the Cascade. When turning off the lamp, please stop at the fan position for a minute to cool the lamp before going to the off position.
----- Original Message -----
From: Woo Shik Jung<alsune at stanford.edu>
To: cascade at snf.stanford.edu
Cc: Jim McVittie<mcvittie at cis.stanford.edu>
Sent: Thu, 23 Jan 2014 15:23:10 -0800 (PST)
Subject: Cascade lamp source repair update
Dear cascade users,
We had a malfunctioning of the lamp source yesterday and have found out
with the help of Jim McVitte that one of the triac component on the circuit
board has been damaged beyond repair. We have already ordered the
replacement part and we expect it to arrive around next Monday. Hopefully
we can bring up the system by mid-next week.
cascade mailing list
cascade at snf.stanford.edu
-------------- next part --------------
An HTML attachment was scrubbed...
More information about the cascade