Diffusion Morning Report 11/07/02
tberg at snf.stanford.edu
Thu Nov 7 06:45:34 PST 2002
Here you go ,
Tylan 1 is back together and in TCA clean, Nitride has had coat ready
for test, Poly is going down on Monday for pull and clean.
Tube Name/Number Status Comments THICKNESS-Change at
Tylan 1 Down TCA Investigation-plumbing change-completed, tube
changed,starting TCA clean
Tylan 2 Up
Tylan 3 yellow TCA Investigation
Tylan 4 yellow Uniformity issues-approximately 5%
Tylan 5-BBr3 Up
Tylan 6-POCl3 Up
Tylan 7-Dirty Ox Up TCA Investigation
Tylan 8-SiGe up 6.3/?
Tylan 9-Poly Up 28.4/30
Tylan 10-Nitride Down Pull and Clean again-uniformity issues 0/30
Tylan 11-BPSG-LTO yellow Uniformity issues 13.22/30
Tylan 12-FGA Up
Tystar Up 6.75/30???
Doped Poly Down
TEOS yellow testing again
Tylan 16 N/A
Burn Box Up
GAAS 22 UP
GAAS 23 UP
SiH4 for furnaces Change at 100PSI Currently @ 580 PSI
-------------- next part --------------
An HTML attachment was scrubbed...
More information about the diffusion