mahnaz at stanford.edu
Wed Jun 13 14:36:18 PDT 2007
We have changed the ellipsoidal mirror and Mario replaced the lamp as
well. Intensity came right at 15mv/cm2 and Jason tested the aligner and
the exposures are
3612 1 um in hard contact 1.1 to 1.3 seconds exposure
3612 1.6 um in hard contact 1.4 to 1.6 seconds exposure.
We are back to our old values.
Please log your exposure on the chart next to the system and comment on
the coral for issues and problems.
More information about the evalign