Cu target broken & target B mount damaged
toadking at stanford.edu
Tue Feb 15 23:27:08 PST 2005
I deposited 140nm Cu on 3 wafers tonight.
Before the deposition, it looked perfect
However, the target's been broken and the mount seems damaged after the
I'm sorry about that.
Can you let me know what I should do?
-------------- next part --------------
An HTML attachment was scrubbed...
More information about the metalica