found broken wafer chips on vat gate valve. cleaned vat valve sealing surface and gate door. pump down main chamber and left load lock at atm and monitor chamber foreline pressure for 2hrs, did not see a pressure rise on foreline. also, turbo pump speed maintained at 598hz. with load lock at atm.