RAITH UDATE2004-08-28 13:00:07: RESTRICTION: NO operations above 10 keV SYSTEM BACK ON LINE
jwc at snf.stanford.edu
Sat Aug 28 13:21:56 PDT 2004
It is my pleasure to announce that the system is back online in normal
functionality. Today's qualification run is well within specifications
with stitching below spec and superb overlay registration. The chip is
in my dry box if you wish to inspect it. 06192003_6.chip QUAL8282004
red sticker on wafer box. 3.3 U 6.2 V position
RESTRICTION: NO operations above 10 keV DO NOT operate above 10 keV.
See below ...
Should there be any problems or EHT trips immediately inform cole loomis
at RAITH USA.
Zeiss should return later this week to check adjustments made on the
system after the system has had several more day to run in.
I will be unavailable and will be away from the Lab returning September
Thank you for your support! Please play nicely together,
-------- Original Message --------
Subject: Problem raith SNF 2004-08-28 13:00:07: NO operations above 10 keV
Date: Sat, 28 Aug. 2004 13:00:08 -0700
From: jwc at snf.stanford.edu
To: raith-pcs at snf.stanford.edu
we wish to allow the FE-gun two weeks of conditioning before we resume regular operations. This is to allow the FE-tip to condition itself and run in properly.
System was tested repeatedly to 30 keV with no EHT trips or vacuum out gassing above normal levels. This is just a precaution to ensure a long bright FE-Gun tip life.
Thank you for your support!
JWC 08282004:1256 hrs
-------------- next part --------------
An HTML attachment was scrubbed...
More information about the raith-pcs