Stanford RAITH 150 back up and available to Users
toecutter4ranger at gmail.com
Tue Nov 13 18:34:59 PST 2012
Stage controller board(s) connection and / or cable connections likely
cause of clicking sounds we were hearing today.
Tested and calibrated motion of Laser Interferometer Stage in X Y Z axis
again and did not encounter any errors.
Sometimes you just have to do this three times until you get it right...
Thanks for Lev Markov at RAITH for last suggestions to check all the cables
and ribbon connectors.
Users are requested to call me immediately if they encounter problems and I
will be staying in Palo Alto tonight.
-------------- next part --------------
An HTML attachment was scrubbed...
More information about the raith-pcs