Ryan Huain Tu
ryantu at stanford.edu
Tue Jul 13 23:20:11 PDT 2004
I will be cancelling my morning session tomorrow b/c Hitachi is down and I
need both machines. It seems though, that there is a "fix" to this Zoom Y
Quoting Luigi Scaccabarozzi <scaccag at stanford.edu>:
> Hi all,
> ryan and I have found problems today in WF alignment. The error is
> "Zoom(Y) exceeds limit". I tried resetting and shutting down computers
> Elphy, but no way. I tried WF align at WD 4.6 to 10 mm and the numbers
> WF correction are the same. It looks like that the Elphy is stuck on
> numbers (zoom X=1.49, zoom Y=1.52).
> I talked to James, and he says to try again tomorrow morning (Ryan is
> next user) after leaving the Elphy OFF for ~20 min (which I just did).
> If it still does not work, tomorrow morning we can call him and/or
> If people from Raith have any idea or suggestion are welcome otherwise
> James plan is to have another Elphy shipped here tomorrow.
> One more thing. I noticed that the option on LEO Tools-> Preferences->
> SEM condition-> device was set to current device instead of POlaroid
> No idea how that could have been switched but check your account. If
> polaroid is not selected, Align writefiled will definitely not work.
> If you have the wrong option, select polaroid, click apply and reboot
> Any help welcome
More information about the raith