Will be Raith be up on Friday morning? YES! Current PLAN -- system coming back on line 1800 Thursday
jwc at snf.stanford.edu
Wed May 5 17:14:15 PDT 2004
Carl Zeiss and RAITH Field Service have just handed the system back to
me for performance testing and LEO column adjustments evaluation.
1. Complete beam drift evaluations this afternoon. Beam drift is
well within specification (spec. 10 nm/min. measured 3 - 5 nm/min.
2. Hand off the system to Gigi and other available RAITH champions
after GiGi overnight and into Thursday morning.
3. I will return to the system at 10 AM tomorrow and hope to complete
qualification testing by late tomorrow afternoon.
4. I hope to release the system back to users at 6 PM tomorrow.
Thank you for your patience and support!
Peter Catrysse wrote:
>I am losing two 4-hour time slots next week (I had Tuesday and Thursday
>morning) as a result of the training class you are giving. I reserved some
>time this Friday morning. How realistic is it that you will be done by then?
>>From: James Conway [mailto:jwc at snf.stanford.edu]
>>Sent: Wednesday, May 05, 2004 1:21 PM
>>To: Peter Catrysse; Raith SNF Mailing list; ebeam at snf.stanford.edu
>>Subject: I have a 6 inch wafer HEADWAY chuck you can check out from me for
>>EBeam Lab use.
>>I have now obtained a 6 inch wafer chuck for the Headway spin coater.
>>This is to be kept secured at my desk and will need to be check out
>>through me for every loan.
>>Thank you for your support!
-------------- next part --------------
An HTML attachment was scrubbed...
More information about the raith