Zoom(Y) out of range
scaccag at stanford.edu
Fri Sep 10 22:37:15 PDT 2004
I guess you are writing at 600X, 100um field. What's the value of zoom Y?
the Y zoom factor must be < ~1.52. you may set 590X, 100um, or decrease
more the mag (570 or 550), that should work, but this is not normal and
its the same problem we had in past. Cole or Joe, can you help with this?
I guess it's again a problem with the calibration of the EO board...
On Fri, 10 Sep 2004, Dong-Woon Shin wrote:
> I am having continuous error in write field alignment - "Zoom(Y) out of
> range: please readjust SEM stage and ELPHY" Only down to 10 um field is OK.
> What does the message mean and what am I supposed to do?
> - DW
> PS. Also I met "access violation err. address 4006A668 in
> module "vcl50.bpl" reads of address 00000060"
More information about the raith