Problem raith SNF 2008-06-03 17:36:17: lots of stage drift 6-2
jwc at snf.stanford.edu
Tue Jun 3 18:46:24 PDT 2008
Hello Kathryn and Raith Users:
Thank you for your report!
I checked the LIS stage today and did not measure any significant drift
or shifting of either the image or the stage SEMing an SOI chip covered
The sample holder may have had a bad landing due to existing issues with
the T loader transfer arm, or your sample could have been charging.
How did the EBL exposure come out?
kathrynt at snf.stanford.edu wrote:
> Image was drifting ~.25 microns/sec Monday night after manual load procedure.
-------------- next part --------------
An HTML attachment was scrubbed...
More information about the raith