savannah chamber reinstall update: bad news i'm afraid
jprovine at stanford.edu
Mon Jan 17 16:28:23 PST 2011
the chamber returned from cleaning today (monday) at 4pm. i was beginning
installation when one of the two substrate heater connection bolts broke.
it had been bent even before we did removed the chamber for cleaning and it
snapped while tightening the nut on reinstall. i've contacted cambridge
nanotech for the fastest and best resolution to this problem. i'll update
as soon as i know anything new. on the plus side the chamber looks
immaculate post cleaning and had no visible residue when given an organics
clean wipe down.
-------------- next part --------------
An HTML attachment was scrubbed...
More information about the savannah