Problem sem4160 SNF 2007-08-08 00:45:44: Sample Holder
jwc at snf.stanford.edu
Wed Aug 8 09:55:00 PDT 2007
Be sure the loading rod is pulled out fully out to its limits in the
load lock chamber. There is a touchy micro-switch that needs to make
before the LOAD Lock exchange door valve will open on switch actuation.
See me if you have more problems
nppatil at snf.stanford.edu wrote:
> Cannot unload sample holder from Specimen Chamber to Specimen Exchange Chamber.
-------------- next part --------------
An HTML attachment was scrubbed...
More information about the sem4160