pruitt at cdr.stanford.edu
Fri Oct 20 18:42:37 PDT 2000
Len suspects that a large number of the shutdowns could have been avoided
if the IG3 ion gage were checked and reset *before* any mouse clicks are
made. i.e. press the EMIS button before you mouse click to clear any
errors. hopefully this will decrease them number of shutdowns.
also, let Len know when you do have to shut it down if you checked
and it still tripped on a mouse click.
More information about the stsetch