Checked monoblock mfc zero and found Argon valve to port on and Nitrogen off. Selected Nitrogen. Set N2 flow to 7.85 liters and controller flowed the setpoint. Switched input to Argon and flow was 5.5 liters the condition the problem report was issued for. Input to N2 port is now on N2.