Woollam is UP
edmyers at stanford.edu
Fri Sep 17 16:07:15 PDT 2010
Sorry to disappoint those lab members who have decided to head to
Mexico instead of making Woollam measurements, but the system is now
up and available.
The ignitor board (not the problem) and new power supply was replaced
(which was the problem). I also replaced the lamp, aligned the beam
and calibrated the system.
I need to thank Woollam for sending the part for trouble shooting and
even more importantly the step by step directions so that even I
could make the repairs.
More information about the woollam